AIPS HELP file for DOVLAMP in 31DEC18
As of Fri Oct 19 17:49:44 2018
DOVLAMP: Procedure to produce ANTAB-style file for phased-VLA
ASDMFILE Full path name to ASDM dir.
CONFIG 0.0 Select this "configuration"
INNAME In/output file name
INCLASS In/output file class
INSEQ In/output file sequence num
INDISK Disk # for in/output file
BIF 0.0 100.0 Lowest IF number 0=>all
EIF 0.0 100.0 Highest IF number 0=>all
CALSOUR FOR FLUX CALIBRATOR ONLY
REFANT Reference antenna
OUTTEXT Directory to put output file
Use: This procedure us usually used by NRAO operations to make
phased-VLA VLBI amplitude calibration files and is not intended for
general usage. However people with phased-VLA data might want to
improve on their amplitude calibration by, for example, flagging the
switched power before running this procedure.
This procedure optionally loads a SDM (VLA) file and then makes
amplitude calibration information for the phased Jansky VLA (observed
after February 2011) when it is used as a VLBI element. The procedure
processes the VLA only data from WIDAR and using the switched power
and optional primary amplitude calibrator to produce a text file that
can be loaded into the TY and GC table of the VLBI data with ANTAB.
Type RUN DOVLAMP to define the DOVLAMP procedure.
ASDMFILE...To load the data automatically from disk, set this.
The full path name to and including the top directory of
the ASDM/BDF data. Two 64-character strings may be used
for long names. The ASDMFILE(2) will be appended to the
non-blank portion of ASDMFILE(1) to generate the name
used. The code will also manage when ASDMFILE(1) begins
with either '$xxxx/' or 'xxxx:' where xxxx is a logical
(environment) variable. The filename sent to OBIT will
be the translation of $xxxx followed by a '/' followed by
an remaining non-blank characters in ASDMFILE(1) followed
If ASDMFILE is set then CONFIG, INNAME, INCLASS and INSEQ
must be set.
If ASDMFILE is set and INCLASS='TYAPL' it is assumed that
the data is loaded and ASDMFILE will be ignored.
CONFIG.....ASDM files often contain more than one "configuration"
numbered from 0 to N. Configurations may differ in
observing band, number of spectral windows, number of
spectral channels in the spectral windows, etc. Use
BDFLIST to see what configurations are in your data set.
DOVLAMP can load only one configuration at a time.
INNAME.....The name of the file to be processed. If ASDMFILE is set
then this will be the output name.
INCLASS....The class of the file to be processed. If ASDMFILE is set
then this will be the output class. If INCLASS='TYAPL'
then it is assumed that the data has already been loaded
and TYAPL has been run so the data will not be loaded
even if ASDMFILE is set, and TYAPL will not run it again.
INSEQ......The sequence number of the file to be processed. Zero
defaults to the highest sequence number of a file
matching INNAME, INCLASS, and INDISK. If ASDMFILE is
set then this will be the output sequence.
INDISK.....The disk drive number of the file to be corrected.
BIF........First IF to copy with UVCOP then process. 0=>all.
EIF........Highest IF to copy. 0=>all higher than BIF
CALSOUR....Primary flux calibrator, if present in observations, if
left blank a primary flux calibrator will be searched for
in the SU table. If CALSOUR(1)='NOCAL', then a
calibrator will not be searched for.
OUTTEXT....Directory to put output file. Path or environmental
variable will work. The name of the output file is
automatically generated. The path must be shorter than
27 characters (including final /).
DOVLAMP: Procedure to create a-priori calibration for the
Documentor: Amy Mioduszewski
Related Programs: VLAMP, CALIB, CLCAL, CALRD, TYAPL
This procedure is intended to produce the amplitude calibration for
the phased VLA (after February 2011) when it is used as a VLBI
element. The VLA data can be loaded from disk, if needed. It
produces a text file that can be loaded with ANTAB.
The main task used in DOVLAMP is VLAMP, which takes the switched power
converts that to the system temperature (Tsys) and then takes the
harmonic sum of the Tsys over all the antennas multiplied by the
amplitude in the first CL table. The amplitude in the first CL table
has the gain curve polynomial in it. The "Gain" (which really isn't
the gain because some of that is folded into the harmonic sum of the
Tsys) is the sum of the sensitivities of the antennas in the GC table.
Then VLAMP makes the ANTAB style text file that can be loaded into the
DOVLAMP adds some steps in order to improve the calibration. If a
primary amplitude calibrator was observed (3C286, 3C48, 3C147, 3C138
or 3C295) then it will be used to improve the calibration. This
calibrator can either be specified in CALSOUR or, if CALSOUR is blank,
will be found automatically in the SU table.
If INCLASS='TYAPL' then it is assumed that TYAPL has been run and the
procedure will not run it again. This is useful if there has been an
problem later in the procedure since the run of TYAPL is usually the
longest part of the procedure.
Basic steps of procedure:
0) Perform basic checks on inputs
1) Determine if file needs to be loaded (i.e., is ASDMFILE set)
--Yes, run BFD2AIPS, user INNAME, INCLASS and INSEQ as output name
--No, run rest of script on file set in INNAME, INCLASS and INSEQ
2) If BIF and/or EIF are not equal to 0 then
--run UVCOP to copy BIF-EIF and proceed with UVCOPed file
3) If INCLASS is *not* TYAPL then
--run TYSMO to clip bad points in SY table
From now on everything is done to the TYAPL file:
4) If CALSOUR is not set, search for standard VLA flux calibrator
(3C48, 3C138, 3C147, 3C286 or 3C295) in the SU table.
5) If flux calibrator exists in data then:
--Load model for flux calibrator is one exists in the data
--run SETJY on flux calibrator
--run CALIB on flux calibrator
--zero phases on CALIB solutions (want amp only)
--run CLCAL to apply amp corrections to first CL table
6) Run VLAMP to create ANTAB style file.