AIPS HELP file for UV2TB in 31DEC21
As of Mon Nov 29 14:17:27 2021
UV2TB: Converts UV autocorrelation spectra to tables
INNAME Input UV file name (name)
INCLASS Input UV file name (class)
INSEQ 0.0 9999.0 Input UV file name (seq. #)
INDISK 0.0 9.0 Input UV file disk unit #
SOURCES Source list
QUAL -10.0 Source qualifier -1=>all
CALCODE Calibrator code ' '=>all
TIMERANG Time range to copy
STOKES Stokes type to process.
SELBAND Bandwidth to select (kHz)
SELFREQ Frequency to select (MHz)
FREQID Freq. ID to select.
BIF 0.0 100.0 Lowest IF number 0=>all
EIF 0.0 100.0 Highest IF number 0=>all
BCHAN 0.0 2048.0 Lowest channel number 0=>all
ECHAN 0.0 2048.0 Highest channel number
SUBARRAY 0.0 1000.0 Subarray, 0=>all
DOCALIB -1.0 101.0 > 0 calibrate data & weights
> 99 do NOT calibrate weights
GAINUSE CL (or SN) table to apply
DOPOL -1.0 10.0 If >0 correct polarization.
PDVER PD table to apply (DOPOL>0)
BLVER BL table to apply.
FLAGVER Flag table version
DOBAND -1.0 10.0 If >0 apply bandpass cal.
Method used depends on value
of DOBAND (see HELP file).
BPVER Bandpass table version
SMOOTH Smoothing function. See
HELP SMOOTH for details.
OUTVER 0.0 46655.0 Output AS spectrum table
BADDISK Disks to avoid for scratch
Task: This task writes autocorrelation spectra into a table of
type AS. Each autocorrelation visibility in the input uv
data is written into a separate row of the table.
INNAME.....Input UV file name (name). Standard defaults.
INCLASS....Input UV file name (class). Standard defaults.
INSEQ......Input UV file name (seq. #). 0 => highest.
INDISK.....Disk drive # of input UV file. 0 => any.
SOURCES....Source list. ' ' or '*' = all;
a "-" before a source name means all except ANY
QUAL.......Only sources with a source qualifier number in the
SU table matching QUAL will be used if QUAL is not
CALCODE....Sources may be selected on the basis of the
calibrator code given in the SU table.
' ' => any calibrator code selected
'* ' => any non blank code (cal. only)
'-CAL' => blank codes only (no calibrators)
anything else = calibrator code to select.
NB: The CALCODE test is applied in addition to the
other tests, i.e. SOURCS and QUAL, in the
selection of sources to process.
TIMERANG...Time range of the data to be copied. In order:
Start day, hour, min. sec,
end day, hour, min. sec. Days relative to ref.
STOKES.....The Stokes type to accept:
'I' (' ' => 'I')
If the data is RR,LL,RL,LR the following are
valid: 'RR', 'LL', 'RL', 'LR'
If the data is in the form of IQUV then the
following are valid: 'Q', 'U', 'V'
SELBAND....Bandwidth of data to be selected. If more than
one IF is present SELBAND is the width of the
first IF required. Units = kHz. For data which
contain multiple bandwidths/frequencies the task
will insist that some form of selection be made
by frequency or bandwidth.
SELFREQ....Frequency of data to be selected. If more than
one IF is present SELFREQ is the frequency of the
first IF required. Units = MHz.
FREQID.....Frequency identifier to select (you may determine
which is applicable from the OPTYPE='SCAN' listing
produced by LISTR). If either SELBAND or SELFREQ
are set, their values overide that of FREQID.
However, setting SELBAND and SELFREQ may result in
an ambiguity. In that case, the task will request
that you use FREQID.
BIF........First IF to copy. 0=>all.
EIF........Highest IF to copy. 0=>all higher than BIF
BCHAN......First channel to copy. 0=>all.
ECHAN......Highest channel to copy. 0=>all higher than BIF
SUBARRAY...Subarray number to copy. 0=>all.
DOCALIB....If true (>0), calibrate the data using information in the
specified Cal (CL) table for multi-source or SN table for
single-source data. Also calibrate the weights unless
DOCALIB > 99 (use this for old non-physical weights).
GAINUSE....Version number of the CL table to apply to
multisource files or the SN table for single
source files. 0 => highest.
DOPOL......If > 0 then correct data for instrumental polarization as
represented in the AN or PD table. This correction is
only useful if PCAL has been run or feed polarization
parameters have been otherwise obtained. See HELP DOPOL
for available correction modes: 1 is normal, 2 and 3 are
for VLBI. 1-3 use a PD table if available; 6, 7, 8 are
the same but use the AN (continuum solution) even if a PD
table is present.
PDVER......PD table to apply if PCAL was run with SPECTRAL true and
0 < DOPOL < 6. <= 0 => highest.
BLVER......Version number of the baseline based calibration
(BL) table to appply. <0 => apply no BL table,
0 => highest.
FLAGVER....specifies the version of the flagging table to be
applied. 0 => highest numbered table.
<0 => no flagging to be applied.
DOBAND.....If true (>0) then correct the data for the shape of the
antenna bandpasses using the BP table specified by BPVER.
The correction has five modes:
(a) if DOBAND=1 all entries for an antenna in the table
are averaged together before correcting the data.
(b) if DOBAND=2 the entry nearest in time (including
solution weights) is used to correct the data.
(c) if DOBAND=3 the table entries are interpolated in
time (using solution weights) and the data are then
(d) if DOBAND=4 the entry nearest in time (ignoring
solution weights) is used to correct the data.
(e) if DOBAND=5 the table entries are interpolated in
time (ignoring solution weights) and the data are then
BPVER......Specifies the version of the BP table to be applied
plotted (if APARM(8)=2) or applied (if DOBAND > 0
and APARM(8) .NE. 2). 0 => highest numbered table.
<0 => no bandpass correction to be applied.
SMOOTH.....Specifies the type of spectral smoothing to be applied to
a uv database . The default is not to apply any smoothing.
The elements of SMOOTH are as follows:
SMOOTH(1) = type of smoothing to apply: 0 => no smoothing
To smooth before applying bandpass calibration
1 => Hanning, 2 => Gaussian, 3 => Boxcar, 4 => Sinc
To smooth after applying bandpass calibration
5 => Hanning, 6 => Gaussian, 7 => Boxcar, 8 => Sinc
SMOOTH(2) = the "diameter" of the function, i.e. width
between first nulls of Hanning triangle and sinc
function, FWHM of Gaussian, width of Boxcar. Defaults
(if < 0.1) are 4, 2, 2 and 3 channels for SMOOTH(1) =
1 - 4 and 5 - 8, resp.
SMOOTH(3) = the diameter over which the convolving
function has value - in channels. Defaults: 1,3,1,4
times SMOOTH(2) used when input SMOOTH(3) < net
OUTVER.....Output AS table version number. 0=>new table.
If the table already exists then new data will be
appended to the end of the table.
BADDISK....The disk numbers to avoid for scratch files (sorting